Veterinary Handheld Pulse Oximeter (OXI-60D)
|FOB Price:||Get Latest Price|
|Min. Order:||30 Pieces|
|Production Capacity:||10, 000PCS/Year|
|Payment Terms:||L/C, T/T, Western Union, Paypal, Money Gram|
- Model NO.: OXI-60D
- Type: Veterinary Oximeter
- Transport Package: Carton
- Origin: China
- Classification: Physiological Functions of Diagnosis and Monitoring Equipment
- Trademark: SHINOVA
- Specification: ISO, CE
- HS Code: 90189090
Veterinary Pulse Oximeter can be used in measuring the pulse oxygen saturation and pulse rate through finger. The product is suitable for being used in family, hospital, oxygen bar, community healthcare, physical care in sports (It can be used before or after doing sports and it is not recommended to use the device during the process of having sport) and etc.
A. Operation of the product is simple and convenient.
B. The product is small in volume, light in weight and convenient in carrying.
1. Main performance
A. SpO2 value display
B. Pulse rate value display, bar graph display
C. Pulse waveform display
D. Low-voltage indication: Low-voltage indicator appears before working abnormally which is due to low-voltage
E. Screen brightness can be changed
F. A pulse sound indication
G. With alarm function
H. With SpO2 value and pulse rate value of storage, the stored data can be uploaded to computers
I. It can be connected with an external oximeter probe
J. Real-time data can be transmitted to computers
K. Review function
L. Clock function
2. Main Parameters
A. Measurement of SpO2 Measuring range: 0%~100% Accuracy: When the SpO2 measuring range is 70%~100%, the permission of absolute error is ± 2%; Below 70% unspecified
B. Measurement of pulse rate Measuring range: 30bpm~250bpm
Accuracy: ± 2 bpm or ± 2% (select larger)
C. Resolution SpO2: 1%, Pulse rate: 1bpm.
D. Measurement Performance in Weak Filling Condition: SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%. SpO2 error is ± 4%, pulse rate error is ± 2 bpm or ± 2% (select larger).
E. Resistance to surrounding light:
The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ± 1%.
F. Power supply requirement: 2.6 V DC ~ 3.6V DC.
G. Optical Sensor
Red light (wavelength is 660nm, 6.65mW) Infrared (wavelength is 880nm, 6.75mW)
H. Adjustable alarm range: SpO2: 0%~100% Pulse Rate: 0bpm~254bpm